The DIGIMAM Phantom was developed in cooperation with the physical-technical quality control group of the European Breast Cancer Network. The phantom complies with the European Guidelines for Quality Assurance in Digital Mammography Screening.
Phantom thickness and material chosen to simulate the average breast Contrast Detail analysis tool containing gold discs Check of geometric distortion Quick check of bad columns Rulers to allow distance measurement CNR measurement Reference point for SNR measurement Check for missed tissue at chest wall Inserts to check on dynamic range with low contrast insert:
- In ‘fatty’ tissue
- In ‘normal’ tissue
- In ‘dense’ tissue Sheet for dynamic range evaluation
Specifications are subject to change without notice. Always read product labelling/inserts for complete instructions, warnings and restrictions.